Changing Your Wi, fi Connection.
Charge maximale 181,4 kg (400 lb) Changement d'utilisateur Type.
3 10 scale Section 2: Technical Features.3 load cell cable connection PIN signal name.
Operating temperature range From 10 to 40C (from 50 to 104F) Radio frequency/ protocol.4 GHz 802.11 b/g/n Wi Fi technology.4 GHz ANT wireless technology.4 GHz Bluetooth Smart wireless technology 2 scale Reading and manual Recording Measurements.Then, restart the gwsl calibration sequence starting at step gwsl *important* Cycle power to accept the calibration.General function: Zero adjustment, auto zeroing, auto calibration.3.3 entering THE function menu.4.1 notes During calibration, use standard weights that are not over owners the scale s maximum capacity.Before cycling power To recalibrate owners simply press the change or zero key with no weight on manual the under hook. 0 F 22 Do not change this setting.
Adjust the reading on the display with the test weight value (change and zero keys) that will be used to calibrate the scale.
Press the lock key, 3 is displayed.
Garmin recommends waiting two hours wallhack before using the aimbot device after eating, drinking, exercising, aimbot bathing, or visiting a sauna.3 11 Section 2: Technical Features.4 KEY button function Press the wallhack ON/OFF key to aimbot power on or power off the scale.Working temperature: -10 C 55 C storage, and 0 C 40 C operating.Note: This should rarely need to be changed, so please ensure this change is necessary before proceeding.The device measures the electrical signal after it passes through your body to estimate aspects of your body composition.Wi, fi connection at any time.Press the zero key, 2 is displayed.Press the change key and hold aimbot for 5 seconds.When the battery charger is in use, it is best not to share power with other equipment on that same outlet.Water Percentage : Water percentage wallhack is the percentage of total mass attributed to fluid.In the programming mode, press this key to change the digit selected.0 F 80 Restore Factory Default 00: Do not change 01: Back to original settings 09/ Rev.When span calibration is captured, the scale will quick emit an audible beep and the unit will display the test load. Press and hold the zero key and the OFF / ON key.